ISO 10360-5:2010 specifies acceptance and periodic reverification tests of CMM performance with contacting probing systems and is only applicable to CMMs using any type of contacting probing system, a discrete point probing mode, and spherical or hemispherical stylus tip(s).
It complements ISO 10360-7, which is the module for CMMs with video probing systems, and ISO 10360-2, which is universal, i.e. not probe-type specific.
ISO 10360-5:2010 applies to CMMs supplied with any of the following: single-stylus probing system; multi-stylus probing systems with fixed multiple styli attached to a single probe (e.g. “star” stylus); multiple probing systems such as those with a stylus for each of their probes; systems with articulating probing; stylus and probe changing systems; manual (non-driven) CMMs.
ISO 10360-5:2010 is not applicable to non-contacting probing systems, which require different testing procedures.